Products shown here are concerned with sample extraction from solids and liquids.
Direct desorption/thermal extraction
Direct desorption is a variation on the principle of dynamic headspace that can be used to rapidly screen small quantities of solid or semi-solid materials. Small samples are placed directly into empty TD tubes or via tube liners. These are then heated in a stream of inert gas to sweep volatiles directly from the sample matrix onto the focusing trap of the thermal desorption instrument, for subsequent analysis by GC.
Microchambers are compact, stand-alone test chamber units that allow rapid sampling of vapour-phase organic compounds from a product or material, complementing small-scale chamber testing by third-party laboratories. Microchamber sampling is a type of dynamic headspace sampling and a powerful tool for increasing laboratory productivity. The technique is usually used in combination with sorbent tubes and off-line thermal desorption–GC–MS.
Soil gas sampling
Quantitative purge-and-trap with GC(–MS) is a common approach for the analysis of VOCs and SVOCs in soil, but it involves labour-intensive sample preparation and pre-screening analysis. Soil probes are a complementary approach to purge-and-trap for analysis of VOCs in soil vapour that overcomes many of these issues. Their use supports compliance with ASTM D7758.
Featured product: HiSorb Starter kit
- A flexible alternative to methods such as stir-bar sorptive extraction (SBSE)
- Detection limits are lower than for SPME
- Easier and quicker to use than solvent extraction
- Re-usable probes and tubes minimise cost per sample
- Robust and easy to use with multiple phases available
- Exclusive to Markes International
Solid-phase microextraction (SPME) uses a small fiber coated with absorptive phase(s), to extract volatiles in either headspace or immersive modes. SPME extraction is an equilibrium process dependent on the partitioning of analytes between the sample matrix (either liquid, solid or headspace) and the absorptive phase. Analytes that become absorbed into the SPME phase are subsequently desorbed using a heated injector, before transfer to the GC column.
Headspace (HS) has been widely used for quantitative evaluation of VOCs in solid and liquid samples for many years. Samples are placed into vials, sealed and incubated. Once equilibrium is reached VOCs partition into the headspace at levels that are proportional to their concentration in the original liquid or solid sample. A precise aliquot of headspace vapour is then transferred/injected directly into the GC (traditional methods) or via an electrically-cooled focusing trap for enhanced performance.